This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||18 September 2013|
|PDF File Size:||6.57 Mb|
|ePub File Size:||5.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
Knob Increments or decrements the current value. In this example you will manial a trace to the internal C: Surround the equipment with three to four inches of packing material and prevent the equipment from moving in the carton.
The mains plug shall be inserted only in a socket outlet provided with a protective earth contact. Save the file 1. Page 34 Making a Basic Measurement This chapter is divided into the following sections: Copying a File 1.
The analyzer was shipped from the factory with the Alignments mode set to Auto. This symbol indicates the input power required is AC.
Agilent Technologies E4440A Getting Started Manual
Put a formatted floppy in the A: To print the manual completely, please, download it. Don’t show me this message again. This symbol is used to indicate the OFF position of the power button. Use the keys to display the softkey that lists the character you want.
Viewing Traces 1 and 2 7. Increments or decrements the current value. Don’t have an account?
If the necessary equipment is available, perform the functional tests in the Instrument Messages and Functional Tests document e44440a your analyzer. Installation and Setup Running Internal Alignments Running Internal Alignments Each time the analyzer is powered on, the internal alignment routine runs automatically.
Restores the analyzer to its factory-defined state. EaEaEaEa.
File Menu Functions File Menu Functions This chapter describes how to use the functions located under the front-panel Data storage and retrieval are handled similarly to that of personal computers PCs. Front-panel keys that are white perform an immediate action rather than bringing up a menu.
Initial Manuaal Initial Inspection Inspect the shipping container and the cushioning material for signs of stress.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
The analyzer has three types of preset: EA 3 Hz – If your analyzer is covered by a separate maintenance agreement, please be familiar with its terms. Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: Agilent Technologies makes no warranty of e4440x kind with regard to this material, including but not limited to, the implied warranties of merchantability and fitness for a particular purpose.
D4440a This feature not fully implemented; to be implemented in a future release. The front-panel keys are dark gray, light gray, green, or white in color. Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted e4440z show which key has been selected but the menu is immediately exited when a selection is made.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
E4440a and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer. Making a Basic Measurement This chapter provides information on basic analyzer operation. Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer.
Page 80 Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC manhal information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz e440a files RF limiters Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
The following table describes the four types of spectrum analyzer messages. Then press the desired terminator either a unit softkey, or the Terminate a value that requires a unit-of-measurement. Used on Option Noise Figure personality.
Refer to Figure In Case of Difficulty Returning an Analyzer for Service Returning an Analyzer for Service Service Tag When you are returning an analyzer to Agilent Technologies for service, fill out and attach one of the blue r4440a tags provided at the end of this chapter. You must now press the NOTE You need to use a filename that does not already exist in the current directory.
Initial Inspection Inspect the shipping container and the cushioning material for signs of stress. In this example you are renaming a trace file; press 3. This analyzer does not contain customer serviceable fuses. If the softkeys Preset Factory Preset 2. You can order the service documentation for Agilent spectrum analyzers through your Agilent Sales and Service office.